engineers use statistical process control (SPC) techniques to
help keep a process within control limits. Defects such as a missing component
or an upside-down connector are easy to identify. Measuring an electrical
quantity provides a wider range of results and involves instruments that
require calibration. Most articles about SPC seem to either assume perfect measurements or assume someone else will discuss
In my opinion, information gathered by SPC tools can offer clues
that help determine when an instrument might need recalibration prior to its
scheduled trip to a cal lab. No one wants to reject good products only because
a DVM went out of calibration.
When you measure, say, a
voltage at a test point on many of the same product and plot voltage versus
number of units, you will see a graph similar to the one shown. The graph
includes the X-bar, or average, value for a good product. You can use this type
of SPC graph to determine how closely the measured voltages match the expected
result. I purposely exaggerated the 35 voltage points to make their value
distinct and easy to see.
This graph also includes
information about the distribution of the voltage values around the 20V
average. On its own, the distribution shows nothing abnormal for this small
sample. I artificially set an upper-control-limit (UCL) and a lower-control
limit (LCL). SPC software would calculate the UCL and LCL from gathered data.
In this type of graph, signs of process problems can include:
1. Eight or more sequential
points above or below the average. 2. Seven or more points in an
up or down "ramp." 3. Seven or more sequential
points with the same value. 4. Fourteen or more
sequential points that alternate above or below the average. 5. Any point above or below the
UCL or LCL lines. 6. Two sequential points
above or below Â±2s (standard deviation). 7. Four out of five
sequential points that fall outside Â±s on the same side of the average.
(Ref. 1, 2)
All these characteristics can arise from production variables,
depending on what you measure. For electrical measurements, recurrence of the
first three situations indicates a need to examine instrument measurement
accuracy and consider the need for recalibration.
Lantronix Inc. has expanded its line of controllers for sensor networks with the release of a rugged controller that improves management of automation systems used in a number of industries, including manufacturing, oil and gas, and chemicals.
Inspired by the hooks a parasitic worm uses to penetrate its host's intestines, the Karp Lab has invented a flexible adhesive patch covered with microneedles that adheres well to wet, soft tissues, but doesn't cause damage when removed.
A quick look into the merger of two powerhouse 3D printing OEMs and the new leader in rapid prototyping solutions, Stratasys. The industrial revolution is now led by 3D printing and engineers are given the opportunity to fully maximize their design capabilities, reduce their time-to-market and functionally test prototypes cheaper, faster and easier. Bruce Bradshaw, Director of Marketing in North America, will explore the large product offering and variety of materials that will help CAD designers articulate their product design with actual, physical prototypes. This broadcast will dive deep into technical information including application specific stories from real world customers and their experiences with 3D printing. 3D Printing is