An instrument such as my trusty Simpson 260 volt-ohm-milliammeter doesn't require calibration because I use it for gross measurements. But engineers must calibrate lab instruments such as oscilloscopes, digital voltmeters and signal sources used for precise measurements. When a calibration comes due, an instrument even a PXI or VXI plug-in instrument goes to an accredited calibration lab. (I'll discuss calibrations more in another column.)
Although measuring-equipment manufacturers specify calibration intervals, typically a year or less, engineers might need to calibrate instruments at other important times, particularly:
When measurement values vary from what you routinely observe or when results seem unusual. These types of observations might signal a need for repair or a thorough clean-out of dust and grime followed by a calibration. Remember, components age so older equipment can need more frequent calibration than newer instruments.
When an instrument experiences rough handling or damage. If a pair of pliers falls on your sensitive DVM or source-measure instrument it shouldn't need calibration. But if the DVM drops off a shelf onto your bench, consider calibrating it. (If you send instruments off-site for calibration, pack them carefully and ensure the cal lab does the same.)
When you start a high-value project. You don't want to get deep into a prototype design only to find you've reported inaccurate test values. Of course you won't know you have bad values until someone tries to duplicate your results or a final design doesn't perform as expected. Some companies might recommend calibrating instruments at the end of such a project, too, just to verify the accuracy of your measurements. A cal lab typically provides a calibration report that includes "as received" information. If the instrument performs within spec, the lab won't perform the associated calibration. Labs also report measured "as sent" information so you know the state of your instrument when it left the lab.
When a contract stipulates instrument calibration. A contract between a company and a university research lab could require calibration of instruments at specific times or by a specific lab.
When production-test equipment produces too many false passes (bad products that test well) or false failures (good products that test bad). No company wants angry customers with faulty products or a rework line jammed with good parts awaiting unnecessary repairs and retesting. You can use statistical process-control (SPC) software to help detect false-pass and false-fail results.
As Sir William Thomson, Lord Kelvin, said, "If you cannot measure it, you cannot improve it." So ensure you keep calibrations up to date.
Some humanoid walking robots are also good at running, balancing, and coordinated movements in group settings. Several of our sports robots have won regional or worldwide acclaim in the RoboCup soccer World Cup, or FIRST Robotics competitions. Others include the world's first hockey-playing robot and a trash-talking Scrabble player.
The DDV-IP is a two-wheeled self-balancing robot that can deliver cold beverages to thirsty folks on hot summer days. A wireless RF remote enables manual control of the device beyond the act of self-balancing. All of the features of the DDV-IP result in an effective delivery vehicle while providing entertainment to the user.
Eric Doster of iFixit talks about the most surprising aspect of the Microsoft Surface Pro 3 teardown. In a presentation at Medical Design & Manufacturing Midwest, iFixit gave the Surface Pro 3 a score of one (out of a possible 10) for repairability.
Focus on Fundamentals consists of 45-minute on-line classes that cover a host of technologies. You learn without leaving the comfort of your desk. All classes are taught by subject-matter experts and all are archived. So if you can't attend live, attend at your convenience.