When good shielding techniques cannot keep noise from affecting measurements, the following tips can help:
Use a spectrum analyzer to help identify noise frequencies and amplitudes. Look for harmonics of noise signals and line-power frequencies. This information can help you identify noise sources.
Separate digital and analog grounds. Use a single-point ground or cluster ground connections as close together as possible to eliminate ground loops. Do not cut off the earth ground (green wire) on a power plug or use a three-wire to two-wire power-outlet adapter to break a ground connection (see photo, below). If you cannot eliminate a ground loop, an isolation amplifier on your signal lines can help.
Suspect all connectors — even those from high-quality suppliers. Broken, high-resistance or corroded ground connections cause problems and increase the noise on measurement signals. Do not mix high-power conductors and low-level signal conductors in the same cable or wire bundle. Separate measurement and control signals as much as possible. Check screw-terminal connections to ensure tight contact. Use high-quality cable and connectors.
Keep connectors clean. 3M's Novec contact cleaner removes grease, silicones and dust, and the 407C audio/video head cleaner from MG Chemicals remove oil and dirt. Neither leaves a residue.
Switch-mode power supplies, motor controllers, electric welders and power-switching equipment can radiate noise. Discharge-lighting equipment and fluorescent-light ballasts can create radiated and conducted noise. You might need to replace old ballasts or install an in-line filter such as the Leviton XPF.
Use power-line EMI filters to attenuate noise conducted in and out of circuits. Some filters include a choke on the ground line to prevent it from conducting noise, too. I know of a portable-phone battery charger that caused considerable RFI over a wide area due to a faulty component and a poor design.
Look for coupling between measurement signals and electromagnetic devices such as motors, relay coils and SMPS transformers that have time-varying magnetic fields. Reorient signal leads perpendicular to magnetic fields or move wires away from these fields.
The electrolyte in aluminum capacitors used in power supplies can dry out and allow noise to pass through to sensitive measurement circuits. If you see line-frequency or SMPS switching noise, look for bad capacitors. It might cost less to replace a bad supply than its bad capacitors.
Look for slots in chassis and enclosures that can radiate RFI signals. If possible, decrease the slot width with conductive tape that connects to the conductive chassis.
Use wide-band ferrite cores, or ferrite noise suppressors, on signal cables to attenuate conducted RFI signals.
Enabling the Future is designing prosthetic appendages modeled more like superhero arms and hands than your average static artificial limbs. And they’re doing it through a website and grassroots movement inspired by two men’s design and creation in 2012 of a metal prosthetic for a child in South Africa.
In order to keep an enterprise truly safe from hackers, cyber security has to go all the way down to the device level. Icon Labs is making the point that security has to be built into device components.
Three days after NASA's MAVEN probe reached Mars, India's Mangalyaan probe went into orbit around the red planet. India's first interplanetary mission, and the first successful Mars probe launched by an Asian nation, has a total project cost of nearly $600 million less than MAVEN's.
Focus on Fundamentals consists of 45-minute on-line classes that cover a host of technologies. You learn without leaving the comfort of your desk. All classes are taught by subject-matter experts and all are archived. So if you can't attend live, attend at your convenience.