While designing a data-acquisition system for a government lab, my colleagues and I added an instrumentation amplifier to each of the system's 16 analog inputs. The buyer offered hazy specifications about the low-level signals the lab wanted to measure, so an instrumentation amplifier (in-amp or INA) provided a way to overcome signal problems that might arise later.
Not all data-acquisition systems require an in-amp, but because they amplify small signal differences from transducers and remove large common-mode ac and dc signals in-amps can come in handy. Think of an in-amp as a "gain block" that amplifies signals by an amount you determine. That may sound like a basic op-amp circuit, but an in-amp differs in several ways. First, an in-amp requires only one resistor to establish its gain. Op-amps typically require several carefully matched resistances and resistance ratios. Second, an in-amp exhibits a high common-mode ac/dc rejection ratio, which exceeds that available in most op-amps. Third, an in-amp provides a high input-impedance load, so it works well with many sensors that produce low-level signals that could include an ac or dc component.
The basic in-amp building block comprises three op-amps as shown in the schematic diagram above. In the strain-gauge bridge circuit shown here, one side of the bridge connects to Input 1 and the other side connects to Input 2. As the resistance of the strain gauge changes, the bridge circuit produces a small voltage difference between Inputs 1 and 2. But both inputs also receive a part of the bridge's excitation voltage, V+. Thus, if Input 1 has an applied voltage of 2.000V, Input 2 might have an applied voltage of 2.015V. The 0.015V difference between the two inputs carries the useful strain information. To start, the in-amp precisely and equally amplifies the Input 1 and 2 signals. (In this example, a single external resistor sets the same gain for both amplifiers).
Next, the two amplified signals go to a circuit that subtracts one signal from the other to remove the bridge circuit's 2.000V dc signal common to both inputs. Engineers refer to that as the common-mode voltage because it appears in common at both inputs. An in-amp also can remove common-mode ac signals, perhaps picked up from surrounding circuits, to present a clean signal to an ADC. Note the in-amp converted a differential signal to a single-ended signal. In-amps provide a reference input, so if the need arises you can offset the in-amp's output.
In addition to accurately amplifying low-level signals, monolithic in-amps have other characteristics such as low input bias and low-noise operation. Some in-amps provide fixed gains that users hard wire or set under computer control. So, when you plan your next measurement system, in-amps should get a close look.
There is currently much discussion around the term "platform," which may be preceded by the adjectives "mobile," "wearable," "medical," "healthcare," etc. However, regardless of the platform being discussed, they usually have one key aspect in common: They tend to be wireless. So, why is this one aspect so fairly universal? The answer is convenience.
Everyone has a MEMS story. For most of us it’s probably the airbag that saved our lives or the life of a loved one. Perhaps it’s the tire pressure sensor that alerted us about deflation before we were stranded alone on a dark muddy road.
Bioimimicry is not merely a helpful design tool -- it also encourages designers to think not only about how to solve design problems by imitating nature, but how to make the products, materials, and systems they design more ecologically sound and nature-friendly.
Focus on Fundamentals consists of 45-minute on-line classes that cover a host of technologies. You learn without leaving the comfort of your desk. All classes are taught by subject-matter experts and all are archived. So if you can't attend live, attend at your convenience.